Frequently Asked Question
The accuracy from 60° to 90° is quite bad that we don’t guarantee it anymore in the datasheet. In fact we suggest our customers to use this model only for applications within ±60°.
The theory behind is that this model uses a 90° inclination MEMS. the angle calculation is an arcsin wave and the non-linearity error outside ±60° is too big to be controlled. All sensors using this principle will have the same problem. We do have calibration which will help to get a better corrected output outside ±60°, however we are cautious with the datasheet, that’s why the measuring range is 90° but the accuracy is only guaranteed within ±60°.